10. Reinsert electrode body into transmitter housing and screw in until plastic is flush with the the housing. Do not overtighten as the plastic can be stripped and make removal and replacement of electrode difficult.
11. Calibrate the electrode. If response time seem slow or drifts for dissolved oxygen, or pH electrodes, rinse the tip with distilled water and place in 0.1M HCl solution for up to 20 mins. This will remove any dust or debris that accumulated during storage. Conductivity and ORP electrodes are more durable and can be scrubbed with a soft brush.
If readings remain slow or other issues occur, please reach out to us at [email protected] so we can help.